This item has been shown 72 times.
Thank you for purchasing this Full Colour Type-C Tester from Hangzhou Ruideng Technologies Co., Ltd. Prior to using this product we recommended that you briefly familiarize yourself with these instructions. In order to ensure the correct operation and use of the device. We also advise that you keep these instructions in a safe place for future reference as may be needed. (Please pay attention to the red words in this manual)1. Technical Parameters:
0.96 Inch color IPS display
Voltage measurement range:
Current measurement resolution:
Capacity accumulation range:
Voltage measurement accuracy:
±(0.5‰+20digits at 23℃)
Energy accumulation range:
Current measurement accuracy:
±(1‰+30 digits, at 23℃)
Load impedance range:
Power measurement range:
Temperature measurement range:
Working temperature range:
about 8g (about package 22g)
Quick charge recognition mode：
QC2.0、QC3.0、APPLE 2.4A/2.1A/ 1A/0.5A、Android DCP、SAMSUN
1.TC66 and TC66C difference: TC66 don't have bluetooth board, don't support APP. TC66C have.
2.Because firware is constantly updated, Please refer to the products actually received.
3.When using TC66 to detect the charger, it is necessary to turn on the CC drop down. When using TC66 to detect the charging process of mobile phone, it is necessary to turn off the CC drop down.
4.The switch ON the product is the power supply switch for the system. When in use, set it to ON. If the voltage is less than 3.5v, use Micro to supply power independently and set the switch to OFF.
5.When TC66 USES the double-headed Typec line for the decoy triggering operation, only one side can be used. If it is not bright, please flip the interface.
6.TC66(C) operation,PC soltware, Android APP instruction download you use a iPhone, you can search "TC66C" in APP STORE and download directly
7.Operation and test video if you insert this meter to a PD charger and the display doesn't light, there are two ways:① you need to connect a load which supports PD protocol too,so that the display will work, ② you can choose to use trigger mode, so that the display will light without connecting a load
10.when you download the file and install the PC software, some windowns virus system will warning and take it as virus, please allow this function or closed the virus system, becuase our file and sofware did not have virus, it was design by our company, some virus system don't know this, it will take all unknown things as system, but we can make sure 100% that there is not virus, if you are care , please stopbuying our port
Expansion measurement measurement measurement measurement measurement method
current can pass in one direction
current pass in twodirection
Identifiable fast charge protocol
Supported fast charge protocol
Huawei SCP、SAMSUNG AFC、PD
Huawei SCP、SAMSUNG AFC、PD2.0、PPS
Quick Charge Recognition Interface
Offline Recording Interface
Quick Charge Recognition Interface
Quick Charge Protocol Detection Interface
System Setting Interface
System Information Interface
Simple Measurement Interface
PC software:work on Win7 or above
(PC software can only be connected by Micro-USB cable)
Mobile Phone:Android5.0 or above；IOS8.0 above(device must over iPhone 6)
(Mobile Phone can only be connected to TC66C by Bluetooth)
Special supported/unsupported fast charge protocol
2.TC66(C) can store Voltage/Current data temporarily,and view data on PC software/APPwhen connected.
3.Trigger/decoy operation/PD trigger operation
4.adjust screen brightness
5.rotate screen direction
6.CC line mode select
8.Restore factory settings
9.Check Products series number/Firmware revision number/Boot times/Running time
10.Only show Voltage measurement/Current measurement/Power measurement/Current direction at Simple Measurement Interface
At this interface, press and hold K2 button to enter trigger menu, then Press K2 to page down to choose, after chosen,press K1 to enter sub-menu. At sub-menu, press K1 or K2 to begin trigger or adjust voltage. Press K2 back to last menu. Except for Huawei SCP and PPS, when other protocol exit trigger mode, they will keep the quick charging state, and re-plug the tester to be restored to the state of 5V. The current protocols supporting for trigger are QC2.0, QC3.0, HUWEI FCP, Huawei SCP PD2.0, PPS and SUMSUNG AFC.(This operation will make the type-c port output high voltage, please unplug the load when using this this function)
3.5.2PD trigger Operation
W：Current PD mode
X: Current actual output Voltage
Y：PPS Voltage step value（cannot be chosen under fixed Voltage）
Z： White options mean supported PD Voltage. Green option means current trigger Voltage.
The PD trigger effect is similar to picture3-5-2, press K1 and re-plug interface according to the tips, then press K1/K2 to switch PD Voltage. Press and hold K1 to switch Voltage step value under PPS mode. Press K1/K2 to increase/decrease the Voltage. Press and hold K2 to exit trigger mode. Because single-sided CC cable is commonly used, when connect PD device with Type-C cable, so flip the interface if it does not work.
Press K2 to enter system setting interface.
3.6 System Setting Interface as picture 3-6-1 and 3-6-2.
AA: Delay off screen setting
AB: Screen brightness setting
AC: Temperature unit switch
AD: PC communication switch
AE: Bluetooth switch
AF: Screen Rotation
AG:PD software switch
AH: Language Choice
AI: Restore factory defaults
At this interface, Press and hold K2 button to enter setting menu, press K2 to page down, after selected. press K1 to enter the current data setting. Press and hold K2 button to return to previous menu. Delay off screen time can be set 1-9 minutes or lighting, 0 mean lighting. There are 10 levels screen brightness; Temperature unit can adjust between ℃ and ℉; there is 180-degree rotation for screen; When the CC pull-down is turned on, the screen will light up by default. If you need to measure PD charging, please turn off the CC drop down; Restore factory defaults only restore setting, not clear data group data and offline data.
For CC pull-down:PD stipulate that type-c interface has no output by default. When there is a drop down on the CC line (pin in the Type-c interface), PD charger thinks that the load is connected and then starts the 5V output. Therefore, it is normal for the tester to plug directly into the PD charger and not light. if CC drop down is enabled in the setting of TC66, when you plug TC66, it will automatically brighten the screen after it is set as enabled, but it may affect the normal charging process. If protocol detection and trigger are to be carried out, CC drop down must be enabled, and the back end shall not be connected with any load. If it is set to be off, the charging of PD charger will not brighten the screen, and the charging process will not be affected. Please first plug the tester into the charger and then insert the load.
Press K2 to enter system information interface.3.7 System Information Interface as picture 3-7
AJ: Products series number
AK: Firmware revision number
AL: Boot times
AM: This time running time
Press K2 button to simple measurement interface.3.8 Simple Measurement Interface as 3-9
This interface only show voltage, current, power and current flow direction3.9Abnormal Status Solution
You can press and hold K2 button and power on TC66 under abnormal status to restore factory settings, or you can press and hold K1 and power on TC66 to enter the boot mode, and use PC software to update the firmware.4. PC software installation instruction
Software installation environment requirements: WIN7 and above system, computer with networks.
This software is developed by hangzhou ruideng technology co., LTD., without virus. If the anti-virus software remind, please allow all its functions, otherwise it will affect the normal operation of the software.
Download software installation
4.1.1. Download the file and open it. RunNET_Framework_4.6 to install .net.
4.1.2 After the installation, double-click the RuiDengUSBMeterto open software.Do not edit or delete other files.
4.1.3 For the first time to use, select Driver Installation menu to install driverfirst.
4.1.4. After installation, Connect the testermicro-USBport tocomputer, the computer will automatically install driver, then unplug and plug it again, the prompt on upper left corner indicates that the device has been inserted, then click the “Connect” to use.4.2 Software operation
Connect the tester with computer by micro cable, when show tester plugged, then press online, then the software will show real-time voltage, current and other data information. The right curve chart can be double-clicked to adapt to the size of the window, and the scroll wheel will zoom in and out. Rotating the scroll wheel between the X-axis and Y-axis, the chart will have different scale effects. Right click on the chart to clear data, and export a table or picture. Click offline data and wait for a moment, offline data will enter the table. When offline data is displayed, real-time curve record will be suspended. Below are the recorded values of the two data groups and the voltage of D+D-.
When there is a new firmware, the tester will remind you to upgrade the firmware. Please follow the following steps to upgrade the firmware
4.3.1.Clean up memory and turn off unnecessary software (or restart your computer).
4.3.2. Restart the software, close all updates prompt.
4.3.3. Press and hold K1 to connect tester with computer, waiting for displaying device inserted, turn off pop-up firmware update prompt (don't click online at this time).
4.3.4. Click on the firmware upgrade, and then click upgrade now.
4.4 Software Update
When there is a new PC software,it will remind you to update, and you need to download and install the new software by yourself. The software version corresponding to the current manual is V184.108.40.206
5.Android APP instruction:
Support Android 5.0 and above, and Bluetooth 4.0and only support TC66C version
you can also download the file with the extension APK and install it on the phone. You can get download TC66C adopted BLE bluetooth module, when android 6.0 and above system connect BLE, it needs location permission. So please agree with this permission when install APP. And BLE can’t be connected by system bluetooth, you need to open APP to search and connect
5.2After connection, then click ENTER to enter operation interface as right picture. Click SHARE export all recording data as EXCEL file. (more functions will be updated later, please look forward to it.)
5.3Software update. After open APP, check for new versions automatically, when there is a new version, the software will automatically prompt update, this manual corresponds to the android software version for V1.1.1.
6. iPhone APP instruction
Support IOS 8.0 and above system(device must over iPhone 6)
The APP of iPhone APP has been approved and searched for "TC66C" to download in APP store