Microamp Millivoltmeter recording Voltmeter ammeter USSR Polygraph new For Sale
This item has been shown 93 times.
Microamp Millivoltmeter recording Voltmeter ammeter USSR Polygraph new:
Designed to measure and record current and voltage values in DC circuits. The limit of permissible basic error in measuring and recording the measured value is ± 1.5% of the final value of the measurement range. Overall dimensions 230x180x315mm. Portable microamperillivoltmeters Н399 (hereinafter referred to as instruments) are used for measuring and recording current and voltage values in direct current circuits. The operating conditions of the device are ambient temperature from 0 to 50 ° С, relative air humidity up to 80% at a temperature of 25 ° С. Н399 microampermillivoltmeters supplied to tropical climatic areas are designed to operate at an ambient temperature of 10 to 45 ° С and relative humidity up to 95% at a temperature of 40 ° С and indoors without sharp changes in temperature, exposure to solar radiation, rain and dust . Perhaps the action of molds and other biological factors. In this case, the factory designation of the device must be supplemented with lettersundefined "04.2". Technical data The margin of permissible basic error in measuring and recording the measured value is ± 1.5% of the final value of the measuring range - for instruments with a zero mark at the edge of the range and of the sum of the final values of the measuring range - for instruments with a zero mark within the measuring range. Final values of measuring ranges: 1mV; 5mV; 10mV; 25mV; 50mV; 75mV; 250mV; 1B; 2.5V; 5V: 10V; 25V; 50V; 100V; 10; fifty; 250 μA. Note. By agreement with the manufacturer, a self-recording Microamperillivoltmeter Н399 / 1 is made on special order with the following final values of the measurement ranges: 5mV; 10mV; 25mV; 50mV; 100mV; 5mkA; 10 μA; 25μA; 50 μA; 100 μA; 10B.
External resistance in the measurement ranges with upper limits from 1mV to 250mV, at which the dynamic properties of the H399 device (quieting, transfer) are guaranteed, no more than 10kOhm External resistance values (R ext) in the measurement ranges with upper limits from 1V to 100V on undefined dynamic properties of the device do not affect. The margin of error of the device for recording time ± 0.5%. The limit of permissible total error of devices for recording the measured value is ± 1.5%. The total error is expressed in the same way as the basic error. The limit of permissible error of the H399 instrument for measuring and recording the measured value caused by a change in the position of the instrument from the normal position in any direction by ± 5 ° is equal to the limit of the permissible basic error. The limit of permissible additional error of the device caused by a change in the supply voltage by ± 10% of the nominal value or the frequency of the supply network by ± 2% of the nominal value is equal to half the value of the limit of the permissible basic error. The limit of permissible additional error of the H399 instruments for measuring and recording the measured value under the influence of an external constant uniform magnetic field with induction of 0.5 mT with the most unfavorable direction of the magnetic field is ± 0.5%. undefined The limit of the permissible additional error of the devices, caused by the installation on a ferromagnetic base with a thickness of (2 +0.5) mm, is equal to half the limit of the permissible basic error. The limit of permissible additional error of the device Microamperillivoltmeter self-recording N399, due to the influence of the same device placed next to it, previously located at a distance of at least 1 m, is ± 0.5%. The limit of the permissible additional error of the H399 instruments for measuring and recording the measured value caused by a change in ambient temperature from normal to any temperature in the range from 0 to 50 ° C is ± 1% for every 10 ° C of temperature change. The insulation between the body of the device and all DC circuits isolated (except for the measuring circuit) can withstand for 1 minute under normal conditions of use. the effect of a test voltage of alternating current with a frequency of 50 Hz, the effective value of which: 2kV - between the network connector and undefined body; 0.5 kV - between the terminals of the timer and the housing; 0.5 kV - between the power terminals from the DC source and the housing. The insulation between separate electrical circuits (except for the measuring circuit), isolated from each other by direct current, withstands normal test conditions, withstands the test voltage: 2kV - between the mains connector and the terminals of the timer; 0.5 kV - between the terminals of the timer and the power terminals from the DC source. Insulation resistance between the housing and DC isolated from the housing by any electrical circuit of the H399 device, except for the measuring one, under normal conditions of use of at least 40 MΩ. The settling time of the moving part of the device does not exceed 2 s. The value of the first flip of the pointer does not exceed 7% of the width of the recording field. The time for establishing the operating mode (self-heating time) is 15 minutes. The readings are recorded in ink on a chart tape in rectangular coordinates. Amount of ink to fit undefined in the ink tank, provides continuous recording of readings on the chart tape for at least 30 days. The width of the working part of the chart tape is 100 mm. The length of the chart tape is not less than 15m. The reading of the readings on the chart tape is carried out using a scale ruler, on which the same marks are applied as on the scale of the H399 device. A synchronous motor powered by AC frequency is used to drive the chart tape.
50Hz, voltage 220V or from a DC source voltage of 12 V through a separate converter P39, supplied on request. The self-recording microamperillivoltmeter H399 has seven speeds for moving the chart tape: 20; 60; 180; 600; 1200; 1800; 5400 mm / h, which are installed by changing a pair of gears. The overall dimensions of the devices are 230x180x315 mm. The mass of the device is not more than 9.0 kg. undefined
We can supply more products, if you need more, make an inquiry?
Items will be shipped from Russia via international registered airmail within 2-3 business days after payment is received and verified. It takes about 14-25 days.
Payment in USA or Europe or Other! WE accept the following payment methods: PayPal, and other.After the sale will be finished we will be to send to you checkout and instruction. Other payments accepteed too please email us.
We do NOT accept returns for complaints from buyers regarding additional items that were not included in the sale description. For example, we will not accept complaints for not including the original box, manuals, or accessoriesIF IT IS NOT STATED IN sale. If you are looking for those specific things when you purchase an item(s), please let us know before you make any purchase so we can work out those details. Thank you.
Vintage Metal Case General Electric Unimeter Type Um-3 With Probes.
Check-point Circuit Tester -2pack
27300 Heavy Duty Circuit Tester With 12' Retractable Wire And Battery Clip
Sg Tool Aid 27250 Circuit Tester With 6-1/2" Long Probe And Retractable Wire